YY/T 0506.4-2016: Translated English of Chinese Standard. (YYT 0506.4-2016, YY/T0506.4-2016, YYT0506.4-2016): Surgical drapes, gowns and clean air suits for patients, clinical staff and equipment - Part 4: Test Method for Linting in the Dry State
This part of YY/T0506 specifies the test method for measuring linting of surgical drapes, gowns, clean air suits and materials under dry state. This method is also applicable for other medical textile materials.
What people are saying - Write a review
We haven't found any reviews in the usual places.
Appendix A Informative General Information on Reproducibility
Appendix B Informative Test Example of the Total Lining of Sample X
air collector Air Sampling Port Appendix applicable arranged average background count C0 bending bonding Box and Air calculate characteristics Clamp classification C0 classification linting Clean Air Suits cleanliness level Close coefficient of linting coefficient of variation diameter disc document equal Figure fixed flow GB/T Gelbo Torsion gives Glue holes Information on Reproducibility ISO Level-5 clean Level-5 clean rooms lg value lined linting on Side-A logarithm materials Maximum measurement channel method motion necessary nonwoven fabrics NOTE number of linting opened operation partial particles particle counter particle size greater period Piece X1 Side-A procedure range rear panel reciprocates Record released selected size classification side Side-A and Side-B similar specified standard deviation starting suitable surface test chamber Test Method Test Piece X1 Test Results torsion device torsional box Total background count total linting Total number Type